53181A LABVIEW DRIVER
If you open the “VI Tree”, you can see all the driver VIs, and also their documentation turn on the “Help” question mark. I have made a Config VI which I have attached. So it takes gates every other measure The code runs through fine and shows no errors until it gets to the Initialise VI and more specifically, the Visa Read section where it is looking for something from the equipment. I am measuring a signal with a gate time of 1 min. Most Active Software Boards:
|Date Added:||24 December 2018|
|File Size:||69.71 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I have made a Config VI which I have attached.
Residual stability is stated at 0. I have an old Fluke that is able to make this measurement BUT it is not gapless. You can also try the shipped examples find it via the Example Findersee the last screenshot below!
I finally found a manual and tried u General Use Programming and Remote Communications clear selected. I’m having difficult in communicating with the frequency counter I’m using.
I have downloaded the driver from here, so I can open your VI now: Hi, I’d appreciate any help in reading data from an HP B. The problem is that you mixed up the proper steps!
Agilent Counter/Timer – IEEE (GPIB) Driver for LabVIEW – National Instruments
Record frequency automatically from 53181A
I have tried many combinations with CALC2: Get a feed of this content Use this view in a tile. I used configure voltage driver to fix the voltage. Looking at the attached driver project, I’ve attached the old and new driver LV Message 1 of 3.
Most Active Software Boards: The code runs through fine and shows no errors until it gets to the Initialise VI and more specifically, the Visa Read section where it is looking for something from the equipment.
One thing I have noticed is that the ‘Remote’ light on the front of the labbiew never illuminates but I’m not sure if its supposed to.
I have also tried to use the provide NI measurement VI and do not get anything from the unit either. I need to make successive second 2 hour period measurement and I want them to be gapless. I use a A frequency counter to measure frequency of a signal.
I’m using labview 8. Log in to follow, share, and participate in this community.
Agilent Technologies / Keysight Technologies 53181A
I finally found the manual and tried some Thanks for the reply. Visa Write shows no errors. Looking at the x driver, you will see i I finally found the manual and tried some of the exa I am measuring a signal with a gate time of 1 min. Message 2 of 3.